You can add Community Subscriptions in the search bar that says "Subscribe to more communities Developed for high-precision radiology and musculoskeletal ultrasound imaging, the ProSound SSD is highly preferred by specialists for its exceptional ease of use, super high-resolution digital imaging mechanism, and versatile compatibility to an extensive line of specialty transducers for enhanced diagnostics. The ProSound SSD was designed with ergonomics, mobility, and user-friendliness to benefit both physicians and patients in mind. Its sophisticated yet simple interface design enables user to easily customize the frequencies and settings to suit specific application. Toggle Navigation. Sign in.
|Published (Last):||13 January 2016|
|PDF File Size:||13.19 Mb|
|ePub File Size:||5.46 Mb|
|Price:||Free* [*Free Regsitration Required]|
Depending on its nature, the issue may be with: 1. Display connect an external display to check 2. UPS you may be using incompatible UPS, try disconnecting it for a few seconds and see if any changes occur 3. Probe try reconnecting, connecting to another port, or checking using another device if possible 4. Device in case these faults are present on all the probes in different ports, the device itself may be faulty.
Try connecting this probe to another port, check it on another device or check if the probe is compatible with this device model. Check if the device is connected to the power source, check replace the power cord, if you work using the UPS, try using the device without it, powering it directly from the mains.
In case you experience any main display issue, try to reconnect its power connectors and signal connectors, or connect it to another source if possible, for example, to a PC. It is necessary to clean the device and check coolers condition, replacing or repairing them if necessary. If you work without UPS: 1. There was an AC power mains failure power surge , in that case, try rebooting the system. The power supply may have been interrupted, check if power is available in the power mains.
In both cases, pay attention what actions cause the system shut down and if any messages errors were returned, in most cases this is due to system overheating, and it is necessary to perform preventive cleaning, etc.
It is necessary to perform hdd system cleaning and diagnostics replacing thermal compound, pads, etc. If none of the keys respond, it is most likely the connector issue, try reconnecting if possible. If only certain keys do not respond, try to disassemble the keyboard and clean it if you have a user manual. It is necessary to clean plugs sockets and also reconnect them. It is also necessary to reconnect the probe, try connecting it to another port, or check it on another device if possible.
If you have a user manual, you can carry out cleaning. If after cleaning, the problem persists, it is most likely a mechanical or electronic failure. If you have a user manual, you can clean the trackball, as it is possible that the gel along with the dust clogs the mechanical part of the trackball during operations. Carefully read the error messages because they are of different complexity from a simple reminder to clean the filter to this or that module failure.
Take a photo if possible. They will help you diagnose a device malfunction and choose the optimal solution. The display rarely fails. As a rule, it occurs due to mechanical shear. A gradual display burnout which leads to reduced contrast and dynamic range is also possible.
If the device does not operate correctly or periodically returns errors, reinstalling the software can fix the issue. Repair Ultrasound Machines. Artifacts or strips appeared on the display. The button sticks or does not respond to input. The device does not recognize the trancduser. The device does not turn on. The device hums, but the display does not turn on. The device started buzzing.
The device turns randomly off. The device works slowly. The keyboard does not respond to input. The probe regularly loses connectivity while working.
The trackball does not respond. The trackball sticks. When turned on, returns an error message. Display repairs. Power connector repairs. Power connectors are the first in line to check if there is an unclear device malfunction. Signal connectors repairs. A rare problem occurring with the sensor connection board or with the sensor connectors.
Software reinstallation. Ventilation and filter system repairs. A very frequent problem, usually caused by filter clogging, a fan can go wrong as well. Share this ad in social networks:. I have read and agree to the terms and conditions. I agree to Receive marketing materials.
Aloka SSD-3500 Service Manual.pdf
Depending on its nature, the issue may be with: 1. Display connect an external display to check 2. UPS you may be using incompatible UPS, try disconnecting it for a few seconds and see if any changes occur 3. Probe try reconnecting, connecting to another port, or checking using another device if possible 4. Device in case these faults are present on all the probes in different ports, the device itself may be faulty. Try connecting this probe to another port, check it on another device or check if the probe is compatible with this device model.
Aloka SSD-5500 - Service manual.pdf
Aloka SSD-3500 User Manual